Hp B Pdf User Manuals. View online or download Hp B Technical Data Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over OPERATING AND SERVICE MANUAL PART NUMBER Instruments Covered by this Manual. HP Model A Signal Generator with .
|Published (Last):||21 October 2009|
|PDF File Size:||5.23 Mb|
|ePub File Size:||9.8 Mb|
|Price:||Free* [*Free Regsitration Required]|
This would leave the Signal Generator in an unknown state. DO NOT recharge or disassemble the battery, and do not dispose of it by burning. Set the spectrum analyzer majual follows: Ouput signal phase is adjustable in 1 degree increments. Indications When a preset is performed, the Signal Generator resets to the following: Typical total FM cfistorfion harmonic and nen-harmonfc is less that 1.
Seal the shipping container securely.
Servicing instructions are for use by service trained personnel only. The following steps mil guide you through the protxss of making the necessary changes to the parameters to meet the specific requirements of this test. Connect the digital multimeter to A13J2 A13J2 pin i.
About this Manual We’ve added 865b manual to the Agilent website in an effort to heip you support your product. The oscilloscope display for nonadjusted Notch lulters is shown in Rgure The key sequence to increment the frequency UP and DOWN may have to 86556b executed several times to stop the signal within 50 mV of ground.
Initialized value and limits of the AM depth increment are as 8566b At times certain events in the frequency switching cycle may be bypassed.
Hewlett-Packard B Signal Generator
However, it is recommended amnual you replace the battery as soon as the error occurs. An instrument manufactured after the printing of this manual may have a serial prefix that is not listed on the title page. Replace the bottom cover by reversing the procedure given in step b. When this is set to position 1, the input impedance is set to 50 ohms; when it is set to position 2, the input impedance is set to high impedance.
Adjustment lb Adjustments Mode! Full text of ” HP: Figure also shows the cable connections for this measurement. The results can cause degradation of device performancet early failure, 8656g immediate destruction. The Signal Generator can only be addressed to listen.
Full text of “HP: B B final”
The highest reading should not exceed the limit specified. Carrier Frequency Center Frequency Accuracy 0. Connect the digital multimeter to the test points manusl in the table below and adjust for a 0. If the carrier frequency increment is set to a value that would cause the Signal Generator to exceed its frequency range, then either the FREQfr or FREQll key is made inactive determined by which frequency limit would be exceeded.
Refer to paragraph for more information about selecting the best mode of Pulse Modulation for your application. Take the difference between dB and the spectrum analyzer reference. Initial Setup The test setup is the same for all of the checks.
Level Flatness kHz to MHz: Connect the frequency counter to J4 using the BNC cable. The level on the spectrum analyzer changes manusl the phase changes.
8656B Synthesized Signal Generator, 0.1 to 990 MHz
Function Vdc Range 20 V 2. DC FM peak deviation Data entries that would be out-of-range for the selected carrier frequency are rejected. The oscilloscope setups are the same as in step Reassemble the Signal Generator by reversing the procedure given in step c and then in step b. Leave the carrier frequency set to MHz. Frequency resolution is 1 Hz.
HP 8656B Manuals
The remaining entries do not need to be changed. Support for Your Product Agilent no longer sells or supports this product. Simultaneous AM or FM at two rates is selected by using internal and external modulation sources. Indications The carrier frequency changes by the 86556b value of frequency increment.
Remove the top cower. Whenever it is likely that the protection has been impaired, the Signal Generator must be made inoperative and be secured against any unintended operation. It will not, however, respond to the Data message unless it has been previously addressed.
Either upper or lower case letters can be used In Data messages, 4.